Nitinol retrieval basket is an instrument used by urologists to retrieve stones from the upper urinary tract. We report the innovative way of using an instrument purposely designed for endourology, by a urological surgeon in endovascular surgery. After an initial failed angioplasty, a 62-year-old man with peripheral vascular disease had a second right groin cross over catheterization angioplasty of the calcifi ed left iliac stenotic disease performed by a radiologist. This procedure was complicated by the fracture of the catheter tip in the external iliac. With no snares available at the time of the second angioplasty, a Urology Consultant successfully used a Zerotip 2.4 F Å~ 120 cm Nitinol Stone Retrieval Basket, avoiding an open laparotomy.
Costin Chintea, Ajantha Jayatunga, Alexander Wolinski, Henryk Krasnowski
Submitted April 16, 2011 - Accepted for Publication May 11, 2011
KEYWORDS: Stone retrieval basket; Urology; Endourology; Angioplasty
CORRESPONDENCE: Costin Chintea, 53 Stonehouse Lane, Quinton, Birmingham, West Midlands B32 3DU, United Kingdom ().
CITATION: Urotoday Int J. 2011 Aug;4(4):art4. doi:10.3834/uij.1944-5784.2011.08.08
Transluminal iliac angioplasty is a very useful and common procedure for treating suitable lesions. Complications of this procedure are uncommon.
Urology is at the forefront of surgery with innovative technology. We describe a case of fragmentation of the angioplasty catheter and an innovative method of retrieving the fragment by an urologist using urological instruments.
A 62-year-old plumber presented with 2-year intermittent claudication and rest pain at night in both his legs. He had very calcifi ed vessels in the groin and pelvis with absent pulses below the femoral arteries. MRA and Duplex scan showed significant stenosis of both common iliac arteries [LCIA & RCIA], left external iliac artery [LEIA] and right superficial femoral artery (RSFA).
Initial management with conservative measures and left groin angioplasty failed. Subsequently, a right groin angioplasty with 8 mm balloon dilatation of the calcifi ed iliac stenotic disease and cross over catheterization was performed through a 6F sheath under heparin cover. Unfortunately, during the procedure the tip of the catheter sheared over the aortic bifurcation and became detached. The dislodged fragment appeared stable and remained impacted in the right external iliac artery. Retrieval of the detached fragment is usually removed using snares but they were unavailable.
A Urology Consultant, available at that time, successfully used an endourological instrument. To start with, the 6F sheath was kept in the RSFA. A Zerotip 2.4 F Å~ 120 cm Nitinol Stone Retrieval Basket was inserted, rotated and moved backwards and forwards under fluoroscopic control. The basket was closed, engaging the tip of the catheter. Once trapped in the basket, the fragment was moved backwards and forward to confirm a good grip on continuous screening. It was then easily retrieved with minimal resistance. Ten milligrams of Alteplase were injected into the artery through the catheter and a good flow was seen within the profunda and popliteal artery. The patient made a satisfactory recovery.
Catheter complications are very rare and occur in 0.6% of cases according to Axisa et al. . Fractured catheter tip is reported in coronary angioplasty [2,3,4] but is very rare in angioplasty of large vessels such as femoral or iliac arteries.
A search of English and Spanish language literature using Medline and PubMed during the period from 1970 to 2011 was performed. The use of a Zero Tip Nitinol Retrieval Basket in vascular surgery has not been reported in the literature.
As an instrument purposely designed for endourology, Zero Tip Nitinol Retrieval Basket features a flat distal surface for atraumatic manipulation. The nitinol wire offers kink-resistance and flexibility. Close stone or catheter tip as in our case proximity could be achieved by the tipless basket configuration.
The use of urological instruments outside urology such as the rigid cystoscope in the anus and rectum has been described previously for the closure of genitourinary fistulae  and the resection of rectal villous adenomas or adenocarcinomas [6,7].
Our case report illustrates the novel use of urological instruments and techniques in endovascular surgery. Urology is a speciality which encompasses a wide variety of skills and its techniques can be easily and safely used for the benefit of other specialities.
In our opinion, these steps can be reproduced and the stone retrieval basket can replace the use of snars in endovascular procedures.
- Axisa B, Loftus I, Bolia A, Fishwick G, Bell PRF, Thompson MM. Complications of percutaneous transluminal angioplasty in the legs: analysis of 1379 procedures. Br J Surg. 2000;87(4):510. CrossRef
- Hwang MH, Hsieh AA, Silverman P, Loeb HS. The fracture, dislodgement and retrieval of a probe III balloon-on-awire catheter. J Invasive Cardiol. 1994;6(5):154–156. PubMed
- Watson LE. Snare loop technique for removal of broken steerable PTCA wire. Cathet Cardiovasc Diagn. 1987;13(1):44–49. PubMed; CrossRef
- Nishiwaki N, Kawano Y, Furukawa K, Nakayama Y. A case report of entrapment of PTCA balloon catheter caused by its rupture. Nippon Kyobu Geka Gakkai Zasshi. 1991;39(8):1226–1230. PubMed
- Ooi J, Stanley J, Lawrentschuk N. Precise application of fibrin glue into a fistula between the rectal stump and urinary bladder—a new technique. The American Journal of Case Reports. 2008;9:178–181.
- Beattie GC, Paul I, Calvert CH. Endoscopic transanal resection of rectal tumours using a urological resectoscope—still has a role in selected patients. Colorectal Dis. 2005;7(1):47–50. PubMed; CrossRef
- Tuech JJ, Pessaux P, Regenet N, Ziani M, Ollier JC, Arnaud JP. Endoscopic transanal resection using the urological resectoscope in the management of patients with rectal villous adenomas. Int J Colorectal Dis. 2004;19(6):569–573. PubMed; CrossRef