This study aims to reduce the uncertainty about the photoneutron dose produced over a course of radiotherapy with high-energy photon beams and evaluate photoneutron contamination-based secondary malignancy risk for different treatment modalities.
FREE DAILY AND WEEKLY NEWSLETTERS OFFERED BY CONTENT OF INTEREST
Did you find this article relevant? Subscribe to UroToday-GUOncToday!
The fields of GU Oncology and Urology are advancing rapidly including new treatments, enrolling clinical trials, screening and surveillance recommendations along with updated guidelines. Join us as one of our subscribers who rely on UroToday as their must-read source for the latest news and data on drugs. Sign up today for blogs, video conversations, conference highlights and abstracts from peer-review publications by disease and condition delivered to your inbox and read on the go.
Dosimetric measurements were taken in Philips SL25/75, Elekta Synergy Platform (Elekta AB, Stockholm, Sweden), Varian Clinac DHX High Performance systems (Varian Medical Systems, Palo Alto, CA), and Cyberknife Robotic Radiosurgery Unit (Accuray Inc., Sunnyvale, CA) using bubble detector for neutron dosimetry. The measurement data were used to determine in-field and out-of-field neutron equivalent dose in 6-MV 3D conformal radiotherapy, sliding window-intensity-modulated radiotherapy, and stereotactic body radiotherapy and to calculate the effective dose in 18-MV 3D conformal radiotherapy and sliding window-intensity-modulated radiotherapy techniques for patients with prostate cancer undergoing a standard treatment. For the 18-MV treatment techniques, the secondary malignancy risk due to the neutron contamination was estimated using the risk factors published by The International Commission on Radiological Protection. The neutron contamination-based secondary malignancy risk for the 18-MV 3D conformal radiotherapy and sliding window-intensity-modulated radiotherapy modalities was found to be 0.44% and 1.45% for Elekta Synergy Platform and 0.92% and 3.0% for the Varian Clinac DHX High Performance, respectively. For 6-MV 3D conformal radiotherapy, sliding window-intensity-modulated radiotherapy, and stereotactic body radiotherapy treatment techniques, neutron equivalent doses inside the treatment field were found to be lower than 40 mSv. Our measurements reveal that equivalent dose and effective dose due to the neutron contamination are at a considerable level for 18-MV sliding window-intensity-modulated radiotherapy treatments, while 6-MV photon beams used in different modalities still induce only negligible photoneutrons. The secondary malignancy risk based on photoneutron should be therefore taken into consideration in case of selecting 18-MV photons in an sliding window-intensity-modulated radiotherapy treatment instead of 6 MV. Technology in cancer research & treatment 2015 Jul 07 [Epub ahead of print]
Fatih Biltekin, Mete Yeginer, Gokhan Ozyigit
Department of Radiation Oncology, Faculty of Medicine, Hacettepe University, Sihhiye, Ankara, Turkey fatih biltekin@hacettepe edu tr , Department of Radiation Oncology, Faculty of Medicine, Hacettepe University, Sihhiye, Ankara, Turkey , Department of Radiation Oncology, Faculty of Medicine, Hacettepe University, Sihhiye, Ankara, Turkey