Comparing the accuracy of MRI/ultrasound-guided target-biopsy by transrectal biopsy (TRB) with elastic versus rigid image fusion versus transperineal biopsy (TPB) with rigid image fusion in a standardized setting.
FREE DAILY AND WEEKLY NEWSLETTERS OFFERED BY CONTENT OF INTEREST
Did you find this article relevant? Subscribe to UroToday-GUOncToday!
The fields of GU Oncology and Urology are advancing rapidly including new treatments, enrolling clinical trials, screening and surveillance recommendations along with updated guidelines. Join us as one of our subscribers who rely on UroToday as their must-read source for the latest news and data on drugs. Sign up today for blogs, video conversations, conference highlights and abstracts from peer-review publications by disease and condition delivered to your inbox and read on the go.
Target-biopsy of six differently sized and located lesions was performed on customized CIRS 070L prostate phantoms. Lesions were only MRI-visible. After prior MRI for lesion location, one targeted biopsy per lesion was obtained by TRB with elastic image fusion with Artemis™ (Eigen, USA), TRB with rigid image fusion with real-time virtual sonography (Hitachi, Japan) and TPB with rigid image fusion with a brachytherapy approach (Elekta, Sweden), each on a phantom of 50, 100 and 150 ml prostate volume. The needle trajectories were marked by contrast agent and detected in a postinterventional MRI.
Overall target detection rate was 79.6% with a slight superiority for the TPB (83.3 vs. 77.8 vs. 77.8%). TRB with elastic image fusion showed the highest overall precision [median distance to lesion center 2.37 mm (0.14-4.18 mm)], independent of prostate volume. Anterior lesions were significantly more precisely hit than transitional and basal lesions (p = 0.034; p = 0.015) with comparable accuracy for TRB with elastic image fusion and TPB. In general, TRB with rigid image fusion was inferior [median 3.15 mm (0.37-10.62 mm)], particularly in small lesions.
All biopsy techniques allow detection of clinically significant tumors with a median error of 2-3 mm. Elastic image fusion appears to be the most precise technique, independent of prostate volume, target size or location.
World journal of urology. 2016 Nov 09 [Epub ahead of print]
N Westhoff, F P Siegel, D Hausmann, M Polednik, J von Hardenberg, M S Michel, M Ritter
Department of Urology, University Medical Center Mannheim, Medical Faculty Mannheim, University of Heidelberg, Theodor-Kutzer-Ufer 1-3, 68167, Mannheim, Germany. ., Department of Urology, University Medical Center Mannheim, Medical Faculty Mannheim, University of Heidelberg, Theodor-Kutzer-Ufer 1-3, 68167, Mannheim, Germany., Department of Clinical Radiology and Nuclear Medicine, University Medical Center Mannheim, Medical Faculty Mannheim, University of Heidelberg, Mannheim, Germany., Department of Radiation Oncology, University Medical Center Mannheim, Medical Faculty Mannheim, University of Heidelberg, Mannheim, Germany.